Know with artron
Easy to Use
Get Pregnant Easier
how to use
- Small box
- Large box
Point the absorbent tip downward. Place the absorbent tip in the urine stream for about 5 seconds to be thoroughly wet.
Otherwise, urine can be collected into a clean cup and the absorbent tip can then be dipped into the cup for about 5-10 seconds until h the urine sample moves into the test window.
Re-cap the device and wait for colored bands to appear and read the results. Read the result in 10 minutes.
Remove the testing device from the foil pouch by tearing it at the notch. Hold the strip at the colored end. Holding the strip vertically, immerse the end of the strip with the arrows into the specimen. Do not immerse past the MAX line.
Take the strip out when the sample has migrated to the test window (about 10 seconds). Lay the strip (MAX side facing up) flat on a clean, dry, non-absorbent surface.
Read the result in 10 minutes, following instructions under the “Result Interpretations” section
What does the result mean?
A pink-colored band appears only in the control region © indicating a negative result for the pregnancy.
A clear pink control band (C) and a detectable test band (T) appear, indicating a positive result for pregnancy.
No visible band in the control region ©. Repeat with a new test device. If the test still fails, please contact the distributor with the lot number.
Canadian Fertility Show 2023
Discover the Latest in Fertility Products with Artron Laboratories at Canadian Fertility Show 2023 Artron Laboratories is excited to announce
Artron Pregnancy Tips: Sperm Health
Artron Pregnancy Tips: Sperm Health Written by Li (Joanna) Tang, MSc., revised by Artron Technical Team Healthy sperm is important
Understand hCG in Early Pregnancy Detection: Insights from Artron’s Knowledge Hub
Medically reviewed by Jun Zhang, Ph.D., written by Joanna Tang, MSc, revised by Cenk Ozkan, MBA. Early pregnancy detection is crucial for ensuring a healthy pregnancy. The sooner
There are no reviews yet.